EZ-3 ECG Thank you for your interest in Futuremed’s EZ-3 ECG. Like other EZ-3 users, we are confident you will enjoy this model ideal for obtaining sharp tracings and accurate reports quickly and easily. EZ-3 features: - One button operation - A rechargeable battery that allows hours of use per charge - Automatic interpretation - Inclusion of measurements and averages This Power Point presentation will show you some of EZ-3’s features and give you an idea of how the electrocardiograph works. Each screen of this presentation will display for about 5 seconds, then automatically advance. To return to a previous screen, use the left arrow on your keyboard. To move ahead, use the right arrow. EZ-3 is a easy to use 12 lead ECG. For about the price of a single channel ECG, EZ-3 gives you pre-formatted 3 channel reports. INSTRUCTIONS FOR LOADING PAPER INTO THE EZ-3 • Drop a pack of z-fold paper in EZ-3’s paper tray. • Pull a couple of sheets out of the tray and close the cover. Paper is now loaded and EZ-3 is ready to use. CONNECTING THE ELECTRODES: • Plug the 10 lead patient cable into the port on the right hand side of EZ-3 (marked EKG/ECG). • Connect each lead wire to an alligator clip adapter. • Place the tab electrodes inside the “teeth” of the alligator clips EZ-3 can also be used with permanent electrodes. Prep patient’s skin and attach the extremity (limb) electrodes to a patient’s arms and legs. The six pre-cordial electrodes are placed on the patient’s chest. Make sure the skin is cleaned prior to lead placement. OPERATION OF EZ-3 In Automatic Mode, EZ-3 will print 12 simultaneous leads in 3 channel format To start recording and printing of EKG, press AUTO. The “MANUAL” key lets you print continuously. Use the UP and DOWN arrows to change the 3 channels being displayed To stop continuous printing, press the “STOP” key. You can set EZ-3 up for your own specific needs (long or short reports, averages, table of measurements, etc.) Programming is as simple as pressing “SHIFT”, a 3 or 4 digit code and “STOP”. For example, you can program EZ-3 to print average leads in a 4 x 3 format. To set EZ-3 up to do this press SHIFT, 1, 2, 6, STOP 6 STOP SHIFT 1 2 EZ-3’s other buttons let you adjust the speed and amplitude of tracings as you need. These buttons adjust wave height (amplitude) These buttons adjust paper speed. The “COPY” key lets you print multiple copies of the same report. EZ-3 has several filters to eliminate AC, baseline and muscle noise. The result is a clean, crisp, preformatted and easy to read report. Although it is compact and portable, EZ-3 can be provided with a variety of cart options. Thank you for taking the time to learn more about EZ-3. If you have any questions please feel free to call us at 1-800-222-6780.